Filed under: Equipment, Guest Blogger, On Location | Tags: Archaeology, canon, capture, dome, guest blogger, lighting array, lithics, Reflectance Transformation Imaging, RTI, stone tools
Our guest blogger is Dr. Leszek Pawlowicz, an Associate Practitioner in the Department of Anthropology, Northern Arizona University, Flagstaff, AZ, USA. He can be contacted at email@example.com. A longer version of this post can be seen at http://rtimage.us/?page_id=27. Thank you, Leszek!
When I learned about Reflectance Transformation Imaging (RTI) back in 2009, one of my first thoughts was that it could be a useful tool for imaging and analyzing lithic archaeological artifacts, flaked stone tools in particular. Not an original thought even back then, and over the next four years I’ve seen the occasional RTI lithic image pop up on the web, demonstrating how useful RTI could be in this application. Early in 2013, I started experimenting with RTI on some modern replica projectile points using Highlight-RTI method. Though I got usable results with these experiments, I decided that Dome-RTI was a more appropriate method because of the reduced data acquisition and processing times.
So began a two-year process of building my first Dome-RTI system and refining it. After multiple iterations of the lighting system, controller, and camera/dome stand, I wound up with an 18″-diameter acrylic dome that produces excellent results and is useful for RTI on larger artifacts. However, it’s grossly over-sized for most of the artifacts I’m interested in documenting. Most flaked stone lithic artifacts in the American Southwest are less than 3 inches in length, and an 18″ dome is easily capable of imaging artifacts of at least 4.5″ in maximum dimension (I’ve gotten useful results on artifacts up to 6″ in length). What’s more, these artifacts are housed in scattered locations (museums, government facilities, universities, etc.), and the large size of the dome and stand make transportation and setup of this big system cumbersome. So, applying lessons learned from the first system, I built a second system with an emphasis on portability and speed (Figure 1):
- Dome diameter is 12″, and sits on a stand that is 13.5″ square; total weight of the dome + stand + camera is less than 4 kg. The small size lets it fit into a Pelican case for easy transport.
- The controller box automatically lights 48 3W LEDs in sequence for the light sources; maximum current is 1 amp, and can be set as low as 150 milliamps. The camera shutter is triggered automatically in sync with the LEDs using either a wired remote cable, an IR remote signal, or a Bluetooth HID transmitter; a manual shutter mode is also available.
- Data acquisition time is about 3 minutes with a Canon S110 camera (12 MP, native 12-bit RAW), about one minute with Canon/Nikon DSLRs. A custom GUI front-end for the PTM and HSH fitters reduces data processing time to 1-3 minutes after the photographs are transferred to a computer.
- Dome is mounted on a hinged stand, which allows artifacts to be swapped in/out in about 10 seconds.
- Entire system is powered by 9-12V DC, either from a wall transformer or appropriate battery power supply.
The system can fit securely in a standard camping backpack with room to spare, with a total weight of less than 5 kg. The option of battery power makes this a truly portable, field-ready RTI system (Figure 2).
When recording archaeological sites out in the field, it is often not possible to collect lithic artifacts to bring back to the lab for proper documentation. You either have to photograph them in the field (usually with less-than-satisfactory resolution of artifact details), or hand-draw the flake scars (a slow and tedious process, and often highly inaccurate). This portable RTI system makes it possible to thoroughly document lithic artifacts on-site.
This system has a few more tricks up its sleeve. Full analysis of a lithic artifact may require microscopic analysis of edgewear to determine how it was used.
A simple reconfiguration of the system (Figure 3) allows high-magnification RTI imaging of lithic artifacts, using either the USB microscope (as pictured), or a DSLR equipped with a macro lens that has a working distance of 6″ or more (roughly 90-100mm focal length). A micrometer stage allows for accurate positioning of the artifact under the microscope.
You can also reconfigure the stand to mount the dome vertically for imaging larger artifacts. While I plan to use it in this mode to image the surface of Southwestern pottery, Figure 4 shows the system in vertical mode being used to image an oil painting.
The normals may be off a bit because of the increased spacing between dome and painting, but you can still get useful results, like the specular mode image shown in Figure 5.
Total parts cost of this portable RTI dome, including the Canon camera, was well under $800. Scaling the dome up to a higher size would increase the expenditure by only the extra cost of the dome plus additional LEDs if desired (e.g. 64 instead of 48). For example, a one-meter dome with 64 LEDs would add approximately $400 to the total cost. In the near future, I hope to post information/instructions online that would allow anyone to build a system of their own. If I can build a system without instructions, I’m sure many others could easily build such a system with instructions.
In an upcoming post, I’ll present some of my lithics RTI imaging results from both of my Dome-RTI systems.
2 Comments so far
Leave a comment